Technology > Calibration
Calibration Array for SWOTSWOT corner reflector deployment in Oklahoma and Texas
Calibration Arrays for NISARNISAR corner reflector deployment in Oklahoma
NISAR corner reflector deployment in Fairbanks and Delta Junction Alaska
2.4 m Trihedral Corner Reflector AssemblyThe following document (15206395-1 Rev A) contains specifications for our current build (August 27, 2022) of an L-band, 2.4 m triangular trihedral corner reflector and has been approved for unlimited release (CL#22-5250). For additional information, please contact Leonard Ortiz (email@example.com)
Rosamond Calibration Array
The Rosamond Corner Reflector Array (RCRA) is located near the south beach of Rosamond Dry Lakebed in California. The array consists of triangular trihedral corner reflectors (CRs):
A typical 2.4-meter Rosamond trihedral corner reflector
- Twenty-three 2.4-meter CRs
- Five 4.8-meter CRs
- Ten 0.7-meter CRs
- The 2.4-meter and 4.8-meter CRs are in a 91° line with respect to each other
- The 4.8-meter CRs look east with an azimuth heading of 350°
- The 0.7-meter CRs are in an 80° line: five have an azimuth heading of 350° and five have a heading of 90°
- Ten of the 2.4-meter CRs look east with an azimuth heading of 350° and the remaining thirteen 2.4-meter CRs look west with an azimuth heading of 170°
Mesh for the 2.4-meter corner reflectors
- 2.4-meters CRs: constructed with Aluminum Alloy 3003 H14 aluminum perforated sheets with staggered round holes of size 0.09375" (2.4 mm).
- 4.8-meters CRs: also constructed with Aluminum Alloy 3003 H14 but with larger 0.5" (12.7mm) round holes, and 48% open surface area. The larger holes in the 4.8-meter CRs allow for lower weight but still compatibility with L-band wavelength.
- 0.7-meters CRs: constructed with 12 (2.053 mm) gauge aluminum solid sheets.
Group photo after assembling a 4.8-meter corner reflector
Group photo after assembling the first 0.7-meter corner reflector
The coordinates listed here are in IGS14 Reference Frame. Please view a chronology of reference frames. The coordinates are expressed as WGS-84 geodetic latitude and height above the reference ellipsoid such that:
- a = 6,378,137 meters
- f = 1/298.257222
These coordinates are modeled as two piece-wise linear functions. The first model was based on multi-year time series analysis of the permanent GPS monument located in Rosamond, CA. Please find further information here. This first model has approximately 2.4 cm/year of plate motion with an azimuth direction of 283°. Specifically the change in East, North, and Vertical before July 6, 2019 is modeled as:
- delta_east_in_meters = -0.023491 * ( gps_time_past_1-jul-2011_in_secs) / 365.25 / 86400
- delta_north_in_meters = 0.005248 * ( gps_time_past_1-jul-2011_in_secs) / 365.25 / 86400
- delta_vertical_in_meters = -0.000670 * ( gps_time_past_1-jul-2011_in_secs) / 365.25 / 86400
After the Ridgecrest M7.1 earthquake on July 6, 2019, the coordinates were adjusted for a one time 1.25 CMS North displacement and with new rates for the North component:
- delta_east_in_meters = -0.023491 * ( gps_time_past_6-jul-2019_in_secs) / 365.25 / 86400
- delta_north_in_meters = 0.008048 * ( gps_time_past_6-jul-2019_in_secs) / 365.25 / 86400
- delta_vertical_in_meters = -0.000670 * ( gps_time_past_6-jul-2019_in_secs) / 365.25 / 86400
L-band image of 350° heading corner reflectors
Accuracy & History
As of November 4, 2015, the location of the bottom vertex of CRs #04 through #12 and CRs #15 through #27 have been surveyed to an accuracy of 2 cm RMS. Several CRs on the west end of the array, specifically CRs #00, #01, #02, #03, #13, and #14 have yet to be surveyed to this accuracy and are believed to be known to around 10 cm RMS.
Due to excessive erosion in the lakebed, CRs #16 and #05 were moved in 2014. CR #16 was moved September 4, 2014 approximately 6.8 meters east of its old location and CR #05 was moved November 4, 2014.
During August 8, 2017 maintenance:
- Found two overturned CRs, #18 and #12.
- #12 was moved ~3 meters north and 2 meters west of its original location.
- #18 was moved ~ 1/4 meter north and ~3 meters west of its original location.
- Sloping elevations required re-adjusted tilts of CRs #11 and #01.
During December 21, 2017 maintenance:
- CRs #00 and #01 were re-aligned and small changes to their positions were necessary. The position of these CRs were remeasured.
- The position of CRs #02, #03, #13 were remeasured.
- Elevation tilts to all the 170 CRs were remeasured.
- As of this time, the position of CR #14 and #15 are still suspect (> 10 CMS), and the tilt of CR #17 is also suspect.
During June 27, 2018 maintenance:
- 0.7-meter CRs #29, #30, #32, #33, #35, #36 were installed on the Lakebed. These CRs are all facing South to accommodate Ka-band calibration.
- Guy wires for CR #20 were re-secured to the ground.
During November, 2018 maintenance:
- Placed PARC at CR#17 location on November 1, 2018 and then repositioned 2.4-meter CR#17 15.6 meters West and 5.8 meters North of its prior location on November 7, 2018.
- Resurveyed 2.4-meter CR#01, CR#02, CR#03, CR#04, and CR#12.
- Removed 0.7-meter CR#33 from Rosamond Dry Lakebed.
During July 25, 2019 maintenance:
- Repositioned 2.4-meter CR#01 5.1 meters East and 1.1 meters North due to excessive Lakebed erosion.
- Resurveyed 2.4-meter CR#00, CR#01, CR#06, CR#07, and CR#08.
- Placed 0.7-meter CR#33 back on Rosamond Dry Lakebed and resurveyed. Orientation is no longer 90 degrees, but rotated for a 350 heading. There are now five 0.7-meter solid corners for a 90 degree heading and five 0.7-meter solid corners for a 350 degree heading.
- Adjusted all remaining corners 1.25 CMS North due to Ridgecrest, CA M 7.1 earthquake on July 6, 2019.
During November 14, 2019 maintenance:
- Repaired 2.4-meter CR#00, remeasured, position changes of -1.6/2.2/0.8 CMS change in E/N/V.
- Removed 0.7 meter CR#28 from the RCRA for maintenance.
During October 14, 2020 maintenance:
- Repaired 2.4 meter CR#02 and re-positioned it 6.5 meters East of its prior location. This target was displaced by weather after January 30, 2020, and is not available in the database after March 14, 2020.
First NISAR 2.8 meter corner reflector installed Oct 21, 2020.
- Placed the first NISAR 2.8 meter corner reflector with a 350 degree heading. Surveyed results in the database can be found with the identifiable label CR#41.
- Placed two movable corner reflector builds for an L-band experiment on the Lakebed. These 2.4 meter corner reflectors have headings 170 and 350 degrees, and the surveyed results can be found in the database with identifiable labels CR#47 and CR#45 respectively. Although these targets are still on the Lakebed until TBD, they will only appear in the database for October 22, 2020.
During June 22, 2021 maintenance:
- Removed the NISAR 2.8 meter CR#41 that was at a 350 degree heading.
- Moved the 170 degree heading 2.4 meter CR#01 approximately 6 meters North and replaced the bottom panel of this CR since it was damaged.
- Reinserted the 350 degree heading 0.7 meter CR#28 back into the RCRA.
During November 10, 2021 maintenance:
- CR#01 was overturned. Re-righted CR#01 and remeasured its position.
- Moved CR#03 ~3 meters West due to erosion and remeasured its position.
- Remeasured the tilts of all the 0.7 m and all the 2.4 m CRs.
During June 29, 2022 maintenance:
Rosamond Lakebed Maintenance June 29, 2022
Relocated overturned CR#01 1.0 meters East and 2.6 meters North.
Re-aligned CR#21 which moved it 9 CMS East and 21 CMS South.
Remeasured the tilts and true headings of all 2.4 m CRs.
During November 29, 2022 maintenance:
Replaced four of the 2.4 meter corner reflectors. These CRs CR#08, CR#11, CR#15, CR#19 are now also elevated to ~15 degrees.
New 2.4 m Rosamond CR #11, mostly West facing
New 2.4 m Rosamond CR #19, mostly East facing
During June 23, 2023 maintenance:
Remeasured the tilt and azimuth of CR#02 and resurveyed CR#27. The new position of CR#27 is < 4 CMS from it original 2014 measurement. Its new position is propagated to its installation date on July 10, 2012, including a south addition of 1.04 cms to account for the Ridgecrest, CA M 7.1 earthquake in July 2019.
Resurveying 4.8 m Rosamond CR #27, mostly East facing
During July 27, 2023 maintenance:
Reconstructed CR#14 but unable to measure new location due to time. Resurveyed CR#23, CR#24, CR#25, CR#26. The new positions of all are < 4 CMS from there original 2014 measurement. Their new positions are propagated to their installation date July 10, 2012, including a south addition of 1.04 cms to account for the Ridgecrest, CA M 7.1 earthquake in July 2019.
Resurveying 4.8 m Rosaomand Corner Reflector
July 27th, 2023, Lakebed Crew
During September 27, 2023 maintenance:
Measured position and tilt of CR#14 and made both retroactive to July 23, 2023, when we moved it 1/2 meters West of its previous location. Reconstructed CR#10 and moved it 4.3 meters West. Reconstructed CR#07 and moved it 2.9 meters West. Both CR#10 and CR#07 were precisely remeasured and their tilts are now 14.6 degrees (CR#10) and 14.7 degrees (CR#07). We found CR#22 which was dislocated by Storm Hillary, however, it was not repairable as it was likely struck by lightning.
CR#22 can be seen overturned in the distance from where it originally was positioned. As a placeholder for this CR, we did position new stands and we anticipate reconstructing this target on our next maintenance trip.
The next maintenance trip will likely be June, 2024.
Corner Reflector (CR) Data
Use the 'Get Corner Reflector Data' button to retrieve the vertices of the the Rosamond corner reflectors (latitude, longitude, orientation, etc) for a particular date. Data goes back to January 1, 2000. Output is a comma-separated value (CSV) file.
Select type tide-free for conventional tide-free crust (ITRF), select type instantaneous-crust for instantanous observed crust, and select type mean-crust for permanent tide. See for example, terrestial reference systems, particulary Figure 1.1 for additional information. The entered data applies the plate motion model as described above. The date also applies the solid tide, pole tide, and ocean tide with the FES2014b Ocean Tide Model and center of mass due to both solid Earth and ocean. The reference system instantaneous-crust is a function of latitude, longitude, and the entered time. The reference system mean-crust is a function of latitude only. See for example, permanent tide deformation.
Get Corner Reflector Data
Use the 'Get Tide Data' button to retrieve 48 hours of the tide perturbations centered at the entered date.
Get Tide Data
Jet Propulsion Laboratory
Mail Stop 300-238
4800 Oak Grove Drive
Pasadena, California, 91109-8099
Tel: (818) 354-0717